In the United States, while the incidence of edentulism continues to decline, rapid population growth coupled with current economic conditions suggest that edentulism and conventional denture use will continue at current or higher numbers. Unfortunately, evidence-based guidelines for the care and maintenance of removable complete denture prostheses do not exist. In 2009, the American College of Prosthodontists (ACP) formed a task force to establish evidence-based
guidelines for the care and maintenance of complete dentures. The task force comprised members of the ACP, the Academy of General Dentistry, American Dental Association (ADA) Council on Scientific Affairs, the American Dental Hygienists’ Association, the National Association of Dental Laboratories, and representatives from GlaxoSmithKline Consumer Healthcare. The review process included the assessment of over 300 abstracts selleck screening library and selection of over 100 articles meeting inclusion criteria check details of this review. The task force reviewed synopses of the literature and formulated 15 evidence-based guidelines for denture care and maintenance. These guidelines were reviewed by clinical experts from the participating organizations and were published in February 2011 issue of The Journal of the American Dental Association for widespread distribution to the dental community. 上海皓元医药股份有限公司 These guidelines reflect the
views of the task force. “
“The aim of this study was to establish the wear and cutting efficiency of tungsten carbide burs from different manufacturers by performing cutting tests with machinable glass ceramic. Cutting tests were performed with 70 tungsten carbide burs from seven manufacturers: (A) Coltene/Whaledent, (B) CEI, (C) Meisinger, (D) Axis,
(E) Komet, (F) Kerr, (G) Edenta. All groups were examined under scanning electron microscope (SEM) before and after the cutting efficiency test for similarities and differences. A specially designed cutting device was used. An electric handpiece was operated at 200,000 rpm with a 120 ml/min coolant water supply rate. The burs were tested under a 165 g constant load using 3 mm wide Macor ceramic as substrate. For each bur the cutting procedure involved a total of five cuts of 3 minutes on every cut, with a total cutting time for each bur of 15 minutes. Data were analyzed using one-way ANOVA at 95.0% confidence level. Significant differences (p < 0.05) were found in the mean cutting rates of the different groups. Groups A and B showed the highest cutting rates. Higher cutting rates were associated with a longer bur lifespan. SEM photomicrographs of the burs and substrates revealed significant changes on the surfaces after the cutting process. The morphology characteristics of tungsten carbide burs are related to their effectiveness.